VA300 treatment instrument principle the micro-dots high-frequency lightning produces a 30000000 Hz ultra high frequency electromagnetic oscillation, with the unique soft silk ( than hair thinner ), at 1/ 100 second time break the hemoglobin, will release the energy transfer to the surface of the skin following removal of lesions, capillaries, and intravascular hemoglobin break make it into small molecules are absorbed by the skin tissue itself, so as to make it reach the disposable permanent eradication of red blood results, each treatment of red blood streak time should not exceed15 minutes, precise digital control system, perfect and concise treatment procedures, simple operation is beneficial to quickly grasp. Two, install note1, instrument placed steadily at room temperature - humidity of not more than 60%10-40, clean operation room or in the treatment of indoor, or will cause the instrument components cannot repair damage.
In 2, a foot switch installation will foot switch plug gaps in alignment apparatus "FOOT SWITCH" interface directly after inserting holes, and the locknut screwing clockwise ( Note: during the tightening process for plug and screw, until cannot insert and tighten, prevent virtual connection ). In 3, a power connection will power line connecting apparatus is directly inserted into the apparatus"220V10A50HZ" interface, and confirm the installation is firm. In 4, the treatment head connecting the treatment head connecting apparatus is inserted directly into the instrument of high frequency interface, clockwise to tighten, and confirm the installation is firm, prevent false contact. 5, connect the power first confirm the emergency stop switch in the up state, whether it be an emergency stop switch clockwise (in the direction indicated by the arrow ) rotation to bounce state, the power supply plug access to good three core power socket, the key is inserted into and rotated clockwise90 degrees to" ON" status; this apparatus electric operation.
1. Vascular removal vascular problems all over the body. Cherry angiomas
2. Benign proliferations skin excrescence:Milia hybrid nevus,intradermal nevus ,flat wart,fat granule
3. Pigments age-spots ,sunspots, hyperpigmentation